- Weight: 1 lb., 3 oz. (.5 kg)
- DryMax-UV overall size: 5.8” wide x 2.8” deep x 2.3” high (14.7 cm x 7.1 cm x 5.8 cm)
- Drying compartment tray: 3.3” x 2.1” x .9” (8.4 cm x 5.3 cm x 2.3 cm)
- Carton size: 6.3” wide, 6.5” high, 3” deep (15.9 cm x 16.5 cm x 7.6 cm)
Hinged lid with a drying compartment that will accommodate all types of hearing aids, including many body worn aids. DryMax UV® by Dry & Store® can also be used with cochlear implant microphones, headpieces, and speech processors (both ear-level and body-worn).
After pressing POWER, DryMax UV will run for 8 hours before automatically turning off.
During the 8-hour cycle, the operating temperature inside DryMax UV is ~98°F (37°C) to 104°F (40°C). If you have rechargeable hearing aids, consider DryBoost UV instead. It has a lower temperature that is suitable for rechargeables.
- Input Power: 100-240 volts AC, 50/60 Hz 0.3A max
- Output: 5 volts ~ 2000mA, micro USB connector–3.65 watts
The DryMax UV is available with US domestic or International power supplies. However, the DryMax UV can be operated on a standard micro USB cable plugged into any USB power outlet. Your DryMax UV can be used anywhere in the world–no voltage converter is necessary. All that’s needed is the appropriate plug adapter. A worldwide configuration is also available that contains a power supply with the following interchangeable blades: AUS, EU, UK, STD.
DryMax UV ships with a six-pack of Dry-Brik Mini® desiccants which is a 6 month supply. The Dry-Brik Mini contains an aggressive sorbent (mol sieve) packaged in an attractive, molded, disposable container. Each Mini has a shelf-life of at least eighteen months, unopened. Once the foil label is removed, the Mini lasts one (1) month. Replacement desiccants are inexpensive and are available from Dry & Store distributors and directly from Ear Technology Corporation. The desiccant also adsorbs odors as it performs the drying process
The UV-C lamp inside the DryMax UV model produces ultraviolet (UV-C) wavelengths of light.
Batteries do not have to be removed from hearing aids before Dry & Store use. We do recommend opening the battery door to allow for better drying in the battery compartment area.
PDFs of instructions in six languages. Click here to download the manual of your choice.
Safety & Regulatory
Certified to International Standard for EMC and product safety & CE marked.
Conforms to IEC 60950-1 Ed. 2.0, CISPR 14-1, Ed. 5.1:2009, CISPR 14-2, Ed. 1.2:2008.
Three years in USA and Canada. Proof of purchase required. Outside of North America, refer to warranty terms provided by distributor.