- Weight: 1 lb., 3 oz. (.5 kg)
- Unit size: 7.2” wide x 4.9” deep x 3.3” high (18.3 cm x 12.4 cm x 8.4 cm)
- Drying compartment tray: 3.9” x 3.8” x 1” (9.9 cm x 9.7 cm x 2.5 cm)
Positive-latching hinged lid with a drying compartment that will accommodate most types of hearing aids, including many body worn aids. Dry & Store Global II can also be used with cochlear implant microphones, headpieces, and speech processors (both ear-level and body-worn).
After pressing the ON button, Dry & Store Global II will run for 8 hours before automatically turning off. However, if the lid is opened before the 8 hour cycle is complete, the unit turns off and the electronics timer resets itself.
- Input Power: 100-240 volts AC, 50/60 Hz, 1.0A maximum.
- Output: 24 VDC at 15W maximum.
The default AC Adapter has a US/North America plug. Global, Global II and Zephyr power supplies are interchangeable. All power supplies are CE marked. They are certified to ETL, UL, and CSA standards. Will also work with an inverter for 12V service. Your Dry & Store Global II can be used anywhere in the world–no voltage converter is necessary. All that’s needed is the appropriate plug adapter. A worldwide configuration is also available that contains a power supply with the following interchangeable blades: AUS, EU, STD, UK.
Global II ships with the custom Dry-Brik® II desiccant. The Dry-Brik contains an aggressive sorbent (mol sieve) packaged in an attractive, molded, disposable container. Each Dry-Brik has a shelf-life of at least eighteen months, unopened. Once the foil label is removed, the Dry-Brik II lasts two (2) months. Replacement desiccants are inexpensive and are available from Dry & Store distributors and directly from Ear Technology Corporation. The desiccant also absorbs odors as it dries.
The Global II lamp produces UV-C wavelengths of light. During its one-and-a-half minute cycle, just 90 seconds, the lamp will generate enough UV-C energy to yield a 99.9% reduction of bacteria, specifically Staphylococcus Aureus.
Batteries do not have to be removed from hearing aids before Dry & Store use. We do recommend the battery doors be open.
Instructions – Operating Manual
Operating instructions are available in 38 languages. Click here to download the manual of your choice.
Safety & Regulatory
Certified to UL & CSA standards. CE marked.
Two years in USA and Canada. Proof of purchase required. Outside of North America, refer to warranty terms provided by distributor.